Solving multi-channel LVDT measurement system cross talk
27 July 2015
When LVDT signal cables are bundled together or laid close to each other in a wire trough, the result can be a mutual interference phenomenon called heterodyning or “beating”.
This effect is created when the frequencies of the oscillator in the individual signal conditioner driving the excitation signal to each LVDT vary slightly.
Since the signal cables are close together, they can capacitively couple among each other, resulting in a low frequency signal that is the difference between the frequencies of the individual oscillators. This different frequency signal can ride on the DC output of the LVDT signal conditioners, appearing as low frequency noise with a period measured in fractions of a second, or as a repetitive slow drift with a period of many seconds. When a pair of LVDTs are used for redundancy, this effect may be evident on only one channel of the system. In systems with larger numbers of LVDT channels, this effect may be found in one or more channels, depending on factors like cable length and layout as well as shielding and grounding.
Typically the way to solve this problem in multiple channel systems is to create a “master” oscillator by using the oscillator from one of the signal conditioners to supply the same excitation frequency to all of the LVDTs. The oscillators in the other LVDT signal conditioners are overridden by the master and considered “slaves”. The idea is that there cannot be any beat frequency interference if all the LVDTs are operating at the exact same frequency.
This practice works quite well but has a major drawback, because the "redundant" system is no longer fully redundant. The system is now dependent on one oscillator in one signal conditioner to supply all the LVDTs with the proper excitation frequency. If the master fails, two things could happen: either your other channels stop operating, or they revert back to their original oscillator frequencies and the multiple channel DC outputs now display beat- frequency-produced noise, ripple, or output drifting. Many companies offer LVDT signal conditioners with a master/slave function, but they are all fully susceptible to the beat frequency problem if the designated master oscillator falters. Sometimes when LVDTs are used in a redundant pair this will be evident on only one channel of the system.
Alliance Sensors Group has considered this beat frequency problem with multiple LVDT installations and developed a successful solution to this problem. As the S1A and its variants have a digital address for RS485 communication capability, we are able to use this digital feature to maintain the full redundancy of multiple LVDT systems. If the "master" were to fail, another "master" having a different digital address would instantly come on stream to maintain a single excitation frequency. In this approach the only thing lost in a "master" oscillator failure is the former master channel itself. For multiple channel LVDT systems, the integrity of the output signals from the other channels is fully maintained. This "auto-mastering" feature is unique to the Alliance Sensors S1A and its variants.
Contact Details and Archive...